INSTALLATION FOR MEASUREMENT OF PARAMETERS OF PULSE PULSEMETER PHOTOMETER
The installation is designed to ensure the calibration of photometers-pulse meters in units of the pulsation coefficient (Kp)
Main technical features
The installation provides a discrete change in the level of illumination. The change in illumination levels provides the following range of ripple coefficients (within): 1-2; 5-10; 40-50 and 90-100%.
The main relative error in determining the actual value of the pulsation coefficient (Kp) does not go beyond ± 5%;